09:02 Feb-20-2017 Re: black light borescopeIn Reply to mike lassman at 14:25 Mar-25-2015 (Opening).
One factor to take into account is attenuation of UV light through the fibres of the borescope.
For UV applications, you may need to have the borescope manufactured with fibres which do not attenuate UV
Alternatively, is there space to run an extra cable alongside of the borescope, with 365nm UV LEDs on the end, so that you generate the UV where you need it, and are then only looking for the green images from defects.
The Research and Applications Development (RAD) group is a newly formed team
within Acuren dedicat
ed to tackling challenging inspection problems. Our focus
is the development of novel, field deployable, advanced inspection techniques
for use in cases where standard NDT methods are ineffective. We don't wait for
new innovations, we engineer them. From concept to commissioning.