where expertise comes together - since 1996 -

# The Largest Open Access Portal of Nondestructive Testing (NDT)

Conference Proceedings, Articles, News, Exhibition, Forum, Network and more

where expertise comes together
- since 1996 -
 1374 views
Technical Discussions
Rio
India, Joined Feb 2017, 3

Rio

India,
Joined Feb 2017
3
10:17 Mar-15-2017
MPI central conductor

What is the equation for current in central confuctor method?

Hamid Reza
,

Hamid Reza

,
Joined Oct 2016
68
16:08 Mar-15-2017
Re: MPI central conductor
In Reply to Rio at 10:17 Mar-15-2017 (Opening).

Dear Rio,

It is a little more complicated than that. Is the conductor in the center or offset? Are you examining a round object? or Is it square or rectangular? Are you examining the outer surface or inner surface? What is the current wave form you are using? What is the ratio of conductor cross section to the part cross section (a.k.a fill factor). How many turns do you want to use (single pass, two pass, etc...)? What is the device you are using for magnetization? (e.g. Magnaflux P70, P1500? it dictates the number of passes). These are all the questions that need to be answered.

1. ASME Sec V, 2015, Page 194, T-754.1 & T754.2

2. ASTM E709, 2015, 12.3.3.2 & X3.2.1.2

3. ASTM E1444, 2016e1, X3.4 >---- important

4. BS EN ISO 9934-1, 2015, 8.3.3.1 & A5

A simple formula to find a ballpark:

I = H x Perimeter

I = Current in ampere (RMS) >----- important to not that this is the RMS of the current waveform!
H = Tangential field strength (Kilo ampere per meter)
Perimeter = Perimeter in millimeters

Calculate the perimeter for outside if you want to inspect the outside and vice versa.

H = Assume that tangential field strength is 2KA/m

I = After finding the RMS value of the current, you have to convert it to the value of the waveform you are using. Refer to BS EN ISO 9934-1, 2015, A.9, Table A1.

Good luck.

Sincerely,
EB

Product Spotlight

#### NEW! The PragmaPro Instrument Platform

The PragmaPro is based on a modular cartridge technology and supports various NDT instrument modal
...
ities such as UT, PAUT, ECT and many more. This new platform is based on a machined, powder-coated aluminum frame for shock-proofness, best sealing qualities and maximum heat dissipation. This is practical to extend the outdoor temperature range and/or to extend the power injected in the transducers. The PragmaPro is aiming at a very wide range of applications, such as weld scanning, corrosion mapping and composite testing.
>

#### SITEX CPSERIES

Teledyne ICM’s CPSERIES has been designed with a view to revolutionizing the handling and perfor
...
mances of portable X-Ray sets. Despite having managed to halve the weight of similar portable X-Ray generators available on the market (while continuing to provide the same power output), the SITEX CPSERIES generators feature a shutter, a laser pointer, a beryllium window, an aluminum filter and two integrated diaphragms (customized sizes are available upon request). Without compromising the robustness and reliability for which ICM products are renowned, the small size and light weight of the SITEX CPSERIES will radically change the way that you perform your RT inspections. And you will see a positive impact in terms of both quality and return on investment (ROI).
>

#### NEW Wheel Type Phased Array Probe

DOPPLER NEW Wheel Type Phased Array Probe, more stable, new tyre makes lesser acoustic attenuation
...
, much lighter makes easier to handle, more slim size, magnetic and mechanical encoder optional etc...more
>

#### ISONIC 3510

Very Powerful Superior Performance Extremely Portable Smart Phased Array Ultrasonic Flaw Detector an
...
d Recorder with 2 Conventional UT and TOFD Channels uniquely combines PA, single- and multi-channel conventional UT, and TOFD modalities providing 100% raw data recording and imaging. Suitable for all kinds of every-day ultrasonic inspections
>

Share...
We use technical and analytics cookies to ensure that we will give you the best experience of our website - More Info
Accept
top
this is debug window