where expertise comes together - since 1996 -

The Largest Open Access Portal of Nondestructive Testing (NDT)

Conference Proceedings, Articles, News, Exhibition, Forum, Network and more

where expertise comes together
- since 1996 -

 700 views
Technical Discussions
wang
wang
07:54 Aug-25-2008
who can tell me this formula's unit?

who can tell me this formula's unit?
N=V/£¨15.16F sin2(¦Ã)), { british standard [BS EN 12668-2:2001]}
N:nearfied length unit=?
V:sound velocity of the circular transducer unit=?
¦Ã:divergence unit=?
F:centre frequency unit=?
15.16: ?

Navita
Navita
00:32 Aug-25-2008
Re: who can tell me this formula's unit?
All dimensions for calculation are in mm for length/ distance, Hz for frequency and seconds for time.

----------- Start Original Message -----------
: who can tell me this formula's unit?
: N=V/£¨15.16F sin2(¦Ã)), { british standard [BS EN 12668-2:2001]}
: N:nearfied length unit=?
: V:sound velocity of the circular transducer unit=?
: ¦Ã:divergence unit=?
: F:centre frequency unit=?
: 15.16: ?
------------ End Original Message ------------

Udo Schlengermann
Consultant, -
Standards Consulting, Germany, Joined Nov 1998, 183

Udo Schlengermann

Consultant, -
Standards Consulting,
Germany,
Joined Nov 1998
183
06:27 Aug-25-2008
Re: who can tell me this formula's unit?
The explanations by Navita on equation A.1 of European Standard EN 12668-2 are correct.

If you use mm/s for velocity and cycles/s = Hz for frequency you get mm for the near field length. The factor 15.16 and sin2(gamma) are dimensionless.

Please note, that the velocity of the block is addressed, not the velocity of the transducer.

Udo Schlengermann
Convener CEN/TC138/WG2

----------- Start Original Message -----------
: who can tell me this formula's unit?
: N=V/£¨15.16F sin2(¦Ã)), { british standard [BS EN 12668-2:2001]}
: N:nearfied length unit=?
: V:sound velocity of the circular transducer unit=?
: ¦Ã:divergence unit=?
: F:centre frequency unit=?
: 15.16: ?
------------ End Original Message ------------

Product Spotlight

PAUT Probes

Typical Phased Array probes have frequencies between 1MHz and 20MHz and the number of wafers is 10
...
to 128. M2 Electronics offers customers conventionally ultrasound probes and the ability to provide high-precision Phased Array Ultrasound Probes of up to 256 wafers. We can also customize the probe for our customers to meet the specific application requirements of the user.
>

Robotic laser shearography enables 100% inspection of complex, flight-critical composite structures

An article in “Composites World Magazine” showcases Non Destructive Testing of aero-structures
...
with Laser Shearography. Over the years Dantec Dynamics has supplied many solutions for the aerospace industry. Referring to specific customer projects several of these cases are examined to outline the advantages of using Laser Shearography for automated defect detection.
>

GEKKO - Portable Phased Array Testing with TFM in Real-Time

The portable phased array testing system GEKKO provides 64 parallel test channels. On creating testi
...
ng parameters the operator is assisted by the CIVA software. Due to its modular set-up the GEKKO instrument is suitable for operators of all skill levels.
>

Portable Ground Penetrating Radar - Proceq GP8000

The Proceq GP8000 portable ground penetrating radar instrument is the beginning of a new era in ND
...
T. The outstanding, patented ultra- wideband technology combined with a compact wireless probe delivers unmatched industry performance. Proceq GP8000 is the world’s first large-scale inspection solution that utilizes Augmented Reality in industrial applications.
>

Share...
We use technical and analytics cookies to ensure that we will give you the best experience of our website - More Info
Accept
top
this is debug window