where expertise comes together - since 1996 -

# The Largest Open Access Portal of Nondestructive Testing (NDT)

Conference Proceedings, Articles, News, Exhibition, Forum, Network and more

where expertise comes together
- since 1996 -

 KARL DEUTSCHINSTRUMENTS AND SYSTEMS FOR NON-DESTRUCTIVE TESTING OF MATERIALS
 612 views
Technical Discussions
Mario Belo
NDT Inspection
Trenes Buenos Aires, Argentina, Joined Jan 2000, 7

Mario Belo

NDT Inspection
Trenes Buenos Aires,
Argentina,
Joined Jan 2000
7
00:43 Mar-08-2001
Attenuation calculation

Dear experts:

To calculate attenuation with angular probes I have to use the Pitch-Catch technique obtaining two echoes(different measures) of the same piece.

Am I proceeding the right way?

Do I have to do this with the control set for 2 probes?

Kind regards.

Mario bello

Udo Schlengermann
Consultant, -
Standards Consulting, Germany, Joined Nov 1998, 181

Udo Schlengermann

Consultant, -
Standards Consulting,
Germany,
Joined Nov 1998
181
00:55 Mar-09-2001
Re: Attenuation calculation
:

Obviously you should use two angle beam probes of same type and angle in separate transmitter receiver mode, i.e. one probe is transmitting, the other is receiving.
Position the probes facing each other on a plate and take an echo at a direct V-path between the probes. Then increase distance till you get a W-path (double V) and take another reading.
The path difference is a single V for the angle used.
So the dB-difference between the two readings give you a measure of attenuation in dB/mm sound travel path.
This technique is not exact, because there are other losses which must be compensated. If both readings are in the farfield of the probes you have to compensate the diffraction losses by 6 dB.
And this technique works only with total reflection, i.e. with angles of 45°. With other angles you get mode conversion and therefore additional losses which are not caused by attenuation.

Kind regards
Udo Schlengermann

: Dear experts:
.
: To calculate attenuation with angular probes I have to use the Pitch-Catch technique obtaining two echoes(different measures) of the same piece.
.
: Am I proceeding the right way?
.
: Do I have to do this with the control set for 2 probes?
.
: Kind regards.
.
:
: Mario bello
.

Product Spotlight

#### A1525 SOLO

A1525 Solo – the most compact and affordable TMF unit with two phased array transducers and 3D v
...
isualization and analysis software in standard delivery set. A compact, ergonomic and easy to handheld Phased Array unit based on Total Focusing Method for easy-going imaging of inspection objects with two-dimensional and three-dimensional visualization and evaluation of inspection results.
>

#### NovaScope 6000

The all-digital Novascope 6000 is a portable, ultra-high precision thickness gauge for high-speed
...
thickness measurement. Novascope 6000 has unmatched capabilities and unique features including: •Superior Resolution with high contrast, high-speed color RF display •High pulser voltage •Real-time video output •Increased internal/external data storage •Programmable SetUp features •Battery & AC Powered
>

#### ISONIC 3510

Very Powerful Superior Performance Extremely Portable Smart Phased Array Ultrasonic Flaw Detector an
...
d Recorder with 2 Conventional UT and TOFD Channels uniquely combines PA, single- and multi-channel conventional UT, and TOFD modalities providing 100% raw data recording and imaging. Suitable for all kinds of every-day ultrasonic inspections
>

#### Research Center IDEKO develops an ultrasonic train wheel inspection system for CAF

The Basque Research centre IK4-IDEKO has developed a state-of-the-art ultrasonic inspection system f
...
or the train wheel. This system secures a sound condition of train wheels and is thus a significant contribution to rail transport safety. The device was delivered to CAF recently and its use allows the manufacturer to become an approved supplier of rolling stock in Italy, as it meets the demanding homologation standards of this country. \\\\r\\\\n
>

Share...
We use technical and analytics cookies to ensure that we will give you the best experience of our website - More Info
Accept
top
this is debug window