where expertise comes together - since 1996 -

# The Largest Open Access Portal of Nondestructive Testing (NDT)

Conference Proceedings, Articles, News, Exhibition, Forum, Network and more

where expertise comes together
- since 1996 -

 663 views
Technical Discussions
Marty Jones
Marty Jones
06:59 Sep-03-1996
Depth of Field

Can you tell me of a good textbook or paper that develops the equations for depth of field of focused transducers?

Robert A. Day
Engineering
Milky Way Jewels, USA, Joined Nov 1998, 40

Robert A. Day

Engineering
Milky Way Jewels,
USA,
Joined Nov 1998
40
09:59 Sep-04-1996
Re: Depth of Field
Marty -

I have good new and bad new. First the bad. There are no
equations for the depth of field of a transducer (that I
know off) and as far as I know there is no definition of
depth of field. The whole concept of depth of field in
optics is acceptable sharpness which is kind hard to get
a grasp on in NDT. I suspect every application is
different and you need to define your terms. I worked on
a program to NDT closure welds for radioactive spent
fuel containers several years ago. I defined acceptable
sharpness as 1 mm. Other applications would use other
numbers.

The good news is that once you have a definition you can
calculate the diffraction field of a transducer and from
that determine the depth of field. The same rules
prevail in ultrasound as optics, i.e. you want the
diameter of the transducer (lens) to be much smaller
than the focal length (large f/number). You generally
want to get depth of field in the part and the focus
changes. Calculating the actual depth of field in the
part is much more difficult even for straight beam. The
depth of field decreases to a value similar to what
would be obtained if the focal length is entirely in the
faster material. This means if you get 2 cm of less than
1 mm focus in water the actual depth of field is about 5
mm. It does vary a bit with ratio of water path to metal
path but because the effective focal length in steel is
shorter and the wavelength increases you wind up with a
shorter depth of field and almost the same focal
diameter (usually a little larger).

Equations are available but solving them is not straight
forward. The following are good starting points:

Exact general solution for right circular piston:
D.G. Crighton, et. al., "Modern Methods in Analytical
Acoustics", 1992, Springer-Verlag, pp 530 - 536.
The talk about the high frequency limit which is the
approximation commonly used.

A more classical and less useful discussion is in
Richardson's "Ultrasonic Physics" on page 43. This is
the usual on axis only solution but he shows his work.

A more accessible but still not for focused transducers
discussion is in Timken's " Elements of Acoustics."
This has the solution in Bessel functions and is more
easily calculated.

There are many approximate solutions for focused
transducers published in the literature, and a few exact
numerical approaches. I have copies of most of them and
am planning to put together a program someday soon.
Moving my office to San Francisco has most of this in
boxes and it's not clear when I will get back to
unpacking them. As soon as I get that organized, I'll
send you a list of papers that discuss this.

Getting the depth of field from these is still not easy
since you still have to decide how to measure focal size.
The old FWHM is good but not always the most appropriate,
do you do 6 or 12dB or do you do something more
sophisticated. Once you have a diffraction code that can
do focused transducers, and have a definition if focal
size, and decide what acceptable focus is, then off
course it's easy.

Hope That helps.

: Can you tell me of a good textbook or paper that develops the equations for depth of field of focused transducers?

W. Grandia
W. Grandia
00:05 Sep-05-1996
Re: Depth of Field

Product Spotlight

#### ISAFE3 Intrinsically Safe Sensor System

ISAFE3 intrinsically safe sensor system of Vallen Systeme is especially targeted at the petrochemica
...
l - as well as oil and gas transportation industry. The sensor system is designed for permanent monitoring or periodic inspection tasks. Sensors are available for different AE-frequency ranges optimized for corrosion and fatigue crack detection and other applications. The ISAFE 3 sensor system consists of an AE-sensor (model ISAS3) certified according to ATEX/IEC for installation in zone 0, gas group IIC, IP68, 20 to +60 °C, and a signal isolator (model SISO3) certified for installation in zone 2. An ISAS3 sensor can be mounted in atmosphere or submerged, e.g. in water or crude oil. It is supported by mounting tools for temporary (magnets) or permanent (welded) installation. ISAFE3 supports automatic sensor coupling test and can be used with any AE signal processor supporting 28V supply at 90 mA peak, e.g. Vallen Systeme ASIP-2/A.
>

#### Combination of Digital Image Correlation and Thermographic Measurements

The combination of measuring results from the digital image correlation (ARAMIS, DIC) and temperat
...
ure measuring data from infrared cameras permits the simultaneous analysis of the thermal and mechanical behavior of test specimens in the materials and components testing field.
>

#### Research and Application Development For NDT

Acuren’s Research and Application Development specializes in the development of advanced ultraso
...
nic inspection techniques and systems for challenging inspection applications, with an emphasis on practical solutions which are field deployable. Services include manual and automated ultrasonic inspection system development, inspection technique optimization using laboratory scale studies and ultrasonic modeling (CIVA, BeamTool), preparing technical justification for technique evaluation and qualification (Probability of Detection and sizing accuracy studies), inspection/calibration/analysis procedure preparation to support field deployment of custom techniques, and development of custom imaging algorithms to support challenging inspection applications.
>

#### NDT Master Lecturer

In the program both university professors and practitioners will give lectures, which guarantees the
...
oretical depth and practical inside. Academic Director: Prof. Dr. Christian Boller
The following lecturers are not complete: Prof. Tadeusz Stepinski, Prof. Wieslaw Staszewski, Prof. Frank Walther, Prof. Giovanni Bruno, Prof. Gerd Dobmann, Prof. Philippe Guy
>

Share...
We use technical and analytics cookies to ensure that we will give you the best experience of our website - More Info
Accept
top
this is debug window