where expertise comes together - since 1996 -

# The Largest Open Access Portal of Nondestructive Testing (NDT)

Conference Proceedings, Articles, News, Exhibition, Forum, Network and more

where expertise comes together
- since 1996 -

 7634 views
Technical Discussions
Angelo Negri
Angelo Negri
00:01 Mar-10-2003
Helium leaks measurement: ppm or mbar*l/s?

In order to perform valves Fugitive Emissions tests, we use a VARIAN sniffer probe helium leak detector, which gives readings in mbar*l/s. Our usual European reference specification (Shell MESC SPE 77/312) also gives limits in the same measurement unit.
First of all, is anybody able to explain the physical meaning of "mbar*l/s"? Is that a flow rate or a concentration?
Then, how is it possible to relate this with "ppm]"? As many leak detectors manufacturers have instruments with both scales, we think there must be a simple relation between the two units.

Johnny Darbonne
Johnny Darbonne
00:54 Apr-22-2005
Re: Helium leaks measurement: ppm or mbar*l/s?
: In order to perform valves Fugitive Emissions tests, we use a VARIAN sniffer probe helium leak detector, which gives readings in mbar*l/s. Our usual European reference specification (Shell MESC SPE 77/312) also gives limits in the same measurement unit.
: First of all, is anybody able to explain the physical meaning of "mbar*l/s"? Is that a flow rate or a concentration?
: Then, how is it possible to relate this with "ppm]"? As many leak detectors manufacturers have instruments with both scales, we think there must be a simple relation between the two units.
: Many thanks in advance, 'NGL
.
mbar*l/s is a flow rate of helium into the mass spectrometer. If you know the flow rate on your sniffer probe(measured using a precision flow measuring device) then you can convert your reading to ppm. 1 ppm = 1X10-6 of total flow. i.e. Your probe has a flow rate of 1-mbar-liter/sec and your mass spectrometer reads 5X10-6 mbar-liters/sec helium then the ppm value is 5 ppm helium.

Product Spotlight

#### NEW Wheel Type Phased Array Probe

DOPPLER NEW Wheel Type Phased Array Probe, more stable, new tyre makes lesser acoustic attenuation
...
, much lighter makes easier to handle, more slim size, magnetic and mechanical encoder optional etc...more
>

#### NDT.net launches mobile-friendly design

NDT.net has revamped its website providing a mobile-friendly design.The front page received a comp
...
letely new design and all other sections are now reacting responsively on mobile devices. This has been a major step to make our website more user- friendly.
>

#### Micro CT

nanoVoxel series has nano-scale resolution upto 500nm by using high voltage X-ray source (from 80-
...
300 kV) with micro/nano focal spot and highly sensitive flat panel detectors and unique Optical coupling detector.
>

#### MUSE Mobile Ultrasonic Equipment

The MUSE, a portable ultrasonic imaging system, was developed for in-field inspections of light-weig
...
ht structures. The MUSE consists of a motor-driven manipulator, a water circulation system for the acoustic coupling and a portable ultrasonic flaw detector (USPC 3010). The MUSE provides images of internal defects (A-, B-,C- and D-scan).
>

Share...
We use technical and analytics cookies to ensure that we will give you the best experience of our website - More Info
Accept
top
this is debug window