where expertise comes together - since 1996 -

# The Largest Open Access Portal of Nondestructive Testing (NDT)

Conference Proceedings, Articles, News, Exhibition, Forum, Network and more

where expertise comes together
- since 1996 -
 808 views
Technical Discussions
Phil Harper
Director, Tribosonics
Tribosonics Ltd, United Kingdom, Joined Feb 2004, 9

Phil Harper

Director, Tribosonics
Tribosonics Ltd,
United Kingdom,
Joined Feb 2004
9
03:16 Feb-17-2004
Piezo crystal centre frequency

Hi,
A rather basic question i fear, but i don't have an answer at the moment.
I have got a piezo-ceramic crystal that is 1mm thick and 16mm diameter. I simply want to work out what the centre frequency of it is?

Any ideas?

Thanks

Paul A. Meyer
R & D,
GE Inspection Technologies, USA, Joined Nov 1998, 47

Paul A. Meyer

R & D,
GE Inspection Technologies,
USA,
Joined Nov 1998
47
05:47 Feb-17-2004
Re: Piezo crystal centre frequency
Hi Phil,
The product of the thickness and the resonant frequency is usually a constant, in this case it is called the thickness frequency constant. PZTs usually have a value around 2000 Hz-m. Dividing this value by the thickness (.001 m) yields the resonant frequency ~ 2 MHz. In you situation, you should identify the material, then obtain the frequency constant from the manufacturer, then perform the above calculation. Because the material can resonate in other directions (other modes), ensure that you ask the manufacturer for the "thickness frequency constant".
Alternatively, you can measure the actual resonance of the piece by plotting its impedance spectrum or inspecting it with an ultrasound instrument, but the calculation is easier.
I hope this helps.
Paul

----------- Start Original Message -----------
: Hi,
: A rather basic question i fear, but i don't have an answer at the moment.
: I have got a piezo-ceramic crystal that is 1mm thick and 16mm diameter. I simply wantto work out what the centre frequency of it is?
: Any ideas?
: Thanks
------------ End Original Message ------------

shan
shan
02:03 Jun-20-2005
Fr Vs G
what happens if the value of G (mS) is very low in a transducer?

Product Spotlight

#### Content on Demand by Eddyfi Technologies

Finding yourself in front of a screen practicing safe social distancing instead of out in the fiel
...
d? At Eddyfi Technologies, we are doing our part in helping you be better prepared for what comes next. We introduce to you Eddyfi on Demand. Fill your schedule with webinars, how-to videos, informative technical papers and our NEW addition: Eddyfi Academy. There, you will have access to FREE online courses and obtain certificates of attendance once completed in order for you to include these e-learning sessions in your log of required training hours. Join us today. Stay safe, stay Beyond Current.
>

#### Ultrasonic Testing Immersion Tanks with Unmatched Scanning Features

TecScan’s non-destructive testing Ultrasonic Immersion Tanks & scanners are designed for high pe
...
rformance and demanding NDT testing applications. Our Scan3D™ line of High Precision Immersion Tanks are specifically designed for automated ultrasonic testing of complex composites parts used in aerospace and industrial applications.
>

#### Compact NDT inspection-heads for measurements with active thermography

The compact inspection head is suitable for thermographic ndt tasks. The uncooled infrared camera
...
is specially developed for NDI-tasks and offers a thermal sensitivity until now known only from thermal imagers with cooled detector. All required components and functions are integrated into the inspection-head. You will only need an ethernet cable to connect the sensor with the evaluation system.
>

#### High-performance Linear Phased Array Probes

Available to order from stock in a range of 5MHz – 7.5MHz and from 16 to 64 elements. Designed w
...
ith piezo-composite elements, Phoenix phased array probes provide high-resolution imaging to maximise sensitivity; accurate ultrasonic detection and sizing of defects in welds; and effective corrosion mapping. Housed in a rugged stainless steel case for on-site industrial NDT applications.
>

Share...
We use technical and analytics cookies to ensure that we will give you the best experience of our website - More Info
Accept
top
this is debug window