where expertise comes together - since 1996 -

# The Largest Open Access Portal of Nondestructive Testing (NDT)

Conference Proceedings, Articles, News, Exhibition, Forum, Network and more

where expertise comes together
- since 1996 -
 811 views
Technical Discussions
Phil Harper
Director, Tribosonics
Tribosonics Ltd, United Kingdom, Joined Feb 2004, 9

Phil Harper

Director, Tribosonics
Tribosonics Ltd,
United Kingdom,
Joined Feb 2004
9
03:16 Feb-17-2004
Piezo crystal centre frequency

Hi,
A rather basic question i fear, but i don't have an answer at the moment.
I have got a piezo-ceramic crystal that is 1mm thick and 16mm diameter. I simply want to work out what the centre frequency of it is?

Any ideas?

Thanks

Paul A. Meyer
R & D,
GE Inspection Technologies, USA, Joined Nov 1998, 47

Paul A. Meyer

R & D,
GE Inspection Technologies,
USA,
Joined Nov 1998
47
05:47 Feb-17-2004
Re: Piezo crystal centre frequency
Hi Phil,
The product of the thickness and the resonant frequency is usually a constant, in this case it is called the thickness frequency constant. PZTs usually have a value around 2000 Hz-m. Dividing this value by the thickness (.001 m) yields the resonant frequency ~ 2 MHz. In you situation, you should identify the material, then obtain the frequency constant from the manufacturer, then perform the above calculation. Because the material can resonate in other directions (other modes), ensure that you ask the manufacturer for the "thickness frequency constant".
Alternatively, you can measure the actual resonance of the piece by plotting its impedance spectrum or inspecting it with an ultrasound instrument, but the calculation is easier.
I hope this helps.
Paul

----------- Start Original Message -----------
: Hi,
: A rather basic question i fear, but i don't have an answer at the moment.
: I have got a piezo-ceramic crystal that is 1mm thick and 16mm diameter. I simply wantto work out what the centre frequency of it is?
: Any ideas?
: Thanks
------------ End Original Message ------------

shan
shan
02:03 Jun-20-2005
Fr Vs G
what happens if the value of G (mS) is very low in a transducer?

Product Spotlight

#### ISONIC 3510

Very Powerful Superior Performance Extremely Portable Smart Phased Array Ultrasonic Flaw Detector an
...
d Recorder with 2 Conventional UT and TOFD Channels uniquely combines PA, single- and multi-channel conventional UT, and TOFD modalities providing 100% raw data recording and imaging. Suitable for all kinds of every-day ultrasonic inspections
>

#### HARDNESS TESTER TKM-459CE combi

TKM-459CE combi applies 2 methods of hardness control: UCI and Leeb. It provides high-accuracy tes
...
ting of metals and alloys as well as items of different sizes and configurations, their hardened layers and galvanic coatings. Device represents results in HB, HRC, HV and others. Shock-, dust- and water-proof housing with intuitive software make this gauge easy to use in all working conditions.
>

#### A1525 SOLO

A1525 Solo – the most compact and affordable TMF unit with two phased array transducers and 3D v
...
isualization and analysis software in standard delivery set. A compact, ergonomic and easy to handheld Phased Array unit based on Total Focusing Method for easy-going imaging of inspection objects with two-dimensional and three-dimensional visualization and evaluation of inspection results.
>

#### Ultrasonic Testing Immersion Tanks with Unmatched Scanning Features

TecScan’s non-destructive testing Ultrasonic Immersion Tanks & scanners are designed for high pe
...
rformance and demanding NDT testing applications. Our Scan3D™ line of High Precision Immersion Tanks are specifically designed for automated ultrasonic testing of complex composites parts used in aerospace and industrial applications.
>

Share...
We use technical and analytics cookies to ensure that we will give you the best experience of our website - More Info
Accept
top
this is debug window