where expertise comes together - since 1996 -

The Largest Open Access Portal of Nondestructive Testing (NDT)

Conference Proceedings, Articles, News, Exhibition, Forum, Network and more

where expertise comes together
- since 1996 -
 509 views
Technical Discussions
Paul H
,
USA, Joined Feb 2020, 3

Paul H

,
USA,
Joined Feb 2020
3
19:49 Feb-19-2020
DAC with Basic Cal Block

Hello,

I am seeking a calculator or instructions on setting up a DAC beyond the first 3 points. I found a drawing awhile back but cant seem to find it. It showed the correct order and each leg and bounces to complete the dac. I will be doing this for a 1" and 1.5" thick block for 45,60, and 70 deg angles. The skip distance will be around 6 inches. Thanks for any help!

Paul H
,
USA, Joined Feb 2020, 3

Paul H

,
USA,
Joined Feb 2020
3
19:52 Feb-19-2020
Re: DAC with Basic Cal Block
In Reply to Paul H at 19:49 Feb-19-2020 (Opening).

.75" and 1.5" blocks sorry not 1"

Paul H
,
USA, Joined Feb 2020, 3

Paul H

,
USA,
Joined Feb 2020
3
20:27 Feb-19-2020
Re: DAC with Basic Cal Block
In Reply to Paul H at 19:49 Feb-19-2020 (Opening).

Per ASME Sec V could i just pick 4 or 5 points that encompass the skip distance and calculate out the surface distance and be ok?

Levi Porter
R & D,
Dacon Inspection Technologies, Thailand, Joined Jan 2017, 56

Levi Porter

R & D,
Dacon Inspection Technologies,
Thailand,
Joined Jan 2017
56
06:56 Feb-27-2020
Re: DAC with Basic Cal Block
In Reply to Paul H at 20:27 Feb-19-2020 .

Trigonometric Info

Hi Paul,
I'll take a stab at helping you since nobody has replied.

Could you give a little more information with regard to your calibration blocks. 0.75" and 1.5" are standard sizes for ASME basic blocks. These usually have both side drilled holes and notches. If you are using Notches to create your TCG and performing inspection with 60 degree and 70 degree refraction in steel, you should remember that mode conversion occurs.

A dac curve is created by plotting the amplitude responses from equal size, shape, and oriented reflectors over different distances. You can plot as many points as wanted / needed as long as you continue to get discernible signals over greater distances.

I think you are trying to calculate the DAC sound path when calibrating on an ASME Basic block that is not the same thickness as your inspection piece. For that you can use simple geometry to calculate your inspection sound path, and use those values to ensure your DAC encompasses that sound path.

You can find the full set of trigonometry information here:

https://www.nde-ed.org/GeneralResources/Formula/AngleBeamFormula/AngleBeamTrig.htm

Product Spotlight

NOVO Armor 15 & NOVO Armor 22

The Armor Kit Contains the NOVO Armor, which provides additional mechanical protection to the NOVO 1
...
5WN & NOVO 22WN Detectors, the Armor Stand and a traveling soft cover. - Newest shock absorbent technology case - Water resistant design - Supports wired & wireless communication - Multiple positioning options - Tripod connection using the Built-in 1/4” threads - Simple Detector battery replacement
>

HARDNESS TESTER TKM-459CE combi

TKM-459CE combi applies 2 methods of hardness control: UCI and Leeb. It provides high-accuracy tes
...
ting of metals and alloys as well as items of different sizes and configurations, their hardened layers and galvanic coatings. Device represents results in HB, HRC, HV and others. Shock-, dust- and water-proof housing with intuitive software make this gauge easy to use in all working conditions.
>

iProbe - USB Phased Array Probe

Turn your PC, Laptop or Tablet into a powerful 32:64 Phased Array system with our USB powered Phas
...
ed Array iProbe and inspection software package. Perform affordable weld inspection and corrosion mapping inspections with this powerful but small system. The probe has a 100V pulser, 8K PRF, dual axis encoder input and achieves outstanding signal quality.
>

NEW! The PragmaPro Instrument Platform

The PragmaPro is based on a modular cartridge technology and supports various NDT instrument modal
...
ities such as UT, PAUT, ECT and many more. This new platform is based on a machined, powder-coated aluminum frame for shock-proofness, best sealing qualities and maximum heat dissipation. This is practical to extend the outdoor temperature range and/or to extend the power injected in the transducers. The PragmaPro is aiming at a very wide range of applications, such as weld scanning, corrosion mapping and composite testing.
>

Share...
We use technical and analytics cookies to ensure that we will give you the best experience of our website - More Info
Accept
top
this is debug window