where expertise comes together - since 1996 -

# The Largest Open Access Portal of Nondestructive Testing (NDT)

Conference Proceedings, Articles, News, Exhibition, Forum, Network and more

where expertise comes together
- since 1996 -

 7661 views
Technical Discussions
Angelo Negri
Angelo Negri
00:01 Mar-10-2003
Helium leaks measurement: ppm or mbar*l/s?

In order to perform valves Fugitive Emissions tests, we use a VARIAN sniffer probe helium leak detector, which gives readings in mbar*l/s. Our usual European reference specification (Shell MESC SPE 77/312) also gives limits in the same measurement unit.
First of all, is anybody able to explain the physical meaning of "mbar*l/s"? Is that a flow rate or a concentration?
Then, how is it possible to relate this with "ppm]"? As many leak detectors manufacturers have instruments with both scales, we think there must be a simple relation between the two units.

Johnny Darbonne
Johnny Darbonne
00:54 Apr-22-2005
Re: Helium leaks measurement: ppm or mbar*l/s?
: In order to perform valves Fugitive Emissions tests, we use a VARIAN sniffer probe helium leak detector, which gives readings in mbar*l/s. Our usual European reference specification (Shell MESC SPE 77/312) also gives limits in the same measurement unit.
: First of all, is anybody able to explain the physical meaning of "mbar*l/s"? Is that a flow rate or a concentration?
: Then, how is it possible to relate this with "ppm]"? As many leak detectors manufacturers have instruments with both scales, we think there must be a simple relation between the two units.
: Many thanks in advance, 'NGL
.
mbar*l/s is a flow rate of helium into the mass spectrometer. If you know the flow rate on your sniffer probe(measured using a precision flow measuring device) then you can convert your reading to ppm. 1 ppm = 1X10-6 of total flow. i.e. Your probe has a flow rate of 1-mbar-liter/sec and your mass spectrometer reads 5X10-6 mbar-liters/sec helium then the ppm value is 5 ppm helium.

Product Spotlight

#### CIVA 2020 UT Module

CIVA NDE Simulation Software is the world leader of NDT Simulation. The UT simulation Module incl
...
udes: - "Beam computation": Beam propagation simulation - "Inspection Simulation": Beam interaction with flaws or specimens The user can simulate a whole inspection process (pulse echo, tandem or TOFD) with a wide range of probes (conventional, Phased- arrays or EMAT), components, and flaws.
>

#### NDT Master Lecturer

In the program both university professors and practitioners will give lectures, which guarantees the
...
oretical depth and practical inside. Academic Director: Prof. Dr. Christian Boller
The following lecturers are not complete: Prof. Tadeusz Stepinski, Prof. Wieslaw Staszewski, Prof. Frank Walther, Prof. Giovanni Bruno, Prof. Gerd Dobmann, Prof. Philippe Guy
>

#### NEW Wheel Type Phased Array Probe

DOPPLER NEW Wheel Type Phased Array Probe, more stable, new tyre makes lesser acoustic attenuation
...
, much lighter makes easier to handle, more slim size, magnetic and mechanical encoder optional etc...more
>

#### FD800 Bench Top Flaw Detectors

The bench-top FD800 flaw detector range combines state-of-the-art flaw detection with advanced mater
...
ial thickness capabilities. Designed for use in the laboratory these gauges are the tool you need for all your flaw detecting needs.
>

Share...
We use technical and analytics cookies to ensure that we will give you the best experience of our website - More Info
Accept
top
this is debug window