where expertise comes together - since 1996 -

# The Largest Open Access Portal of Nondestructive Testing (NDT)

Conference Proceedings, Articles, News, Exhibition, Forum, Network and more

where expertise comes together
- since 1996 -

 ACS-Solutions GmbHThe priority of the company is to design and create highly technological equipment, for professional users. Aiming to reach the devices top techn ...
 2092 views
Technical Discussions
Johnnie
Johnnie
02:22 Jul-10-2005
Ultrasonic angle beam formula for cylindrical forgings.

I would like to know the formula for proper angle use to impinge id of thick wallcylindrical forgings/pipe.

Ed T.
Ed T.
05:25 Jul-13-2005
Re: Ultrasonic angle beam formula for cylindrical forgings.
----------- Start Original Message -----------
: I would like to know the formula for proper angle use to impinge id of thick wallcylindrical forgings/pipe.
------------ End Original Message ------------

SinØ 1 = (ID/OD) * SinØ2
SINØ2 being the desired angle on the ID surface.

K.V. Rasmussen
K.V. Rasmussen
09:15 Jul-13-2005
Re: Ultrasonic angle beam formula for cylindrical forgings.
----------- Start Original Message -----------
: I would like to know the formula for proper angle use to impinge id of thick wallcylindrical forgings/pipe.
------------ End Original Message ------------

Many years ago I found some useful formulas in Materials Evaluation No. 43, November 1985, page 1489.
This was published under the heading 'NDT Solutions' by Jim Johnson.

If you are interested contact me directly via e-mail and I can send you a PDF copy.

N.DHANASEKARAN
R & D
WELDING RESEARCH INSTITUTE, BHEL, India, Joined Jun 2005, 3

N.DHANASEKARAN

R & D
WELDING RESEARCH INSTITUTE, BHEL,
India,
Joined Jun 2005
3
01:42 Jul-15-2005
Re: Ultrasonic angle beam formula for cylindrical forgings.

Hello!
Well known formula to calculate the penetrated depth (d) by angle beam (¥è), knowing outer radius(r) of the pipe is given below:

d = r (1- sin¥è)

By substituting the thickness of the pipe in place of "d", for full penetration by the sound beam, one can calculate the angle required.
For example , 25 mm thick pipe of radius 100 mm will require 45 degree angle probe to detect inner notch in the ref.pipe.
N.DHANASEKARAN

-

---------- Start Original Message -----------
: : I would like to know the formula for proper angle use to impinge id of thick wallcylindrical forgings/pipe.
: Sin¨ª 1 = (ID/OD) * Sin¨ª2
: SIN¨ª2 being the desired angle on the ID surface.
: Sorry misprint.
------------ End Original Message ------------

Product Spotlight

#### NDT.net launches mobile-friendly design

NDT.net has revamped its website providing a mobile-friendly design.The front page received a comp
...
letely new design and all other sections are now reacting responsively on mobile devices. This has been a major step to make our website more user- friendly.
>

#### FMC/TFM

Next generation for Phased Array UT is here now with FMC/TFM! Have higher resolution imaging, impr
...
oved signal to noise ratio, characterize, size and analyze defects better with access to several wave mode views and save raw FMC data for higher quality analysis.  Some of the benefits are:
• Beautiful Image! Easier to understand what you're looking at
• Completely focused in entire image or volume
• Much easier to define setups before inspection
• Easier to decipher geometry echoes from real defects
• Oriented defects (e.g. cracks) are imaged better
• See image from different wave modes from one FMC inspection
• FMC data can be reprocessed/analyzed without going back to the field
>

#### Research Center IDEKO develops an ultrasonic train wheel inspection system for CAF

The Basque Research centre IK4-IDEKO has developed a state-of-the-art ultrasonic inspection system f
...
or the train wheel. This system secures a sound condition of train wheels and is thus a significant contribution to rail transport safety. The device was delivered to CAF recently and its use allows the manufacturer to become an approved supplier of rolling stock in Italy, as it meets the demanding homologation standards of this country. \\\\r\\\\n
>

#### FD800 Bench Top Flaw Detectors

The bench-top FD800 flaw detector range combines state-of-the-art flaw detection with advanced mater
...
ial thickness capabilities. Designed for use in the laboratory these gauges are the tool you need for all your flaw detecting needs.
>

Share...
We use technical and analytics cookies to ensure that we will give you the best experience of our website - More Info
Accept
top
this is debug window