where expertise comes together - since 1996 -

# The Largest Open Access Portal of Nondestructive Testing (NDT)

Conference Proceedings, Articles, News, Exhibition, Forum, Network and more

where expertise comes together
- since 1996 -

 2169 views
Technical Discussions
Johnnie
Johnnie
02:22 Jul-10-2005
Ultrasonic angle beam formula for cylindrical forgings.

I would like to know the formula for proper angle use to impinge id of thick wallcylindrical forgings/pipe.

Ed T.
Ed T.
05:25 Jul-13-2005
Re: Ultrasonic angle beam formula for cylindrical forgings.
----------- Start Original Message -----------
: I would like to know the formula for proper angle use to impinge id of thick wallcylindrical forgings/pipe.
------------ End Original Message ------------

SinØ 1 = (ID/OD) * SinØ2
SINØ2 being the desired angle on the ID surface.

K.V. Rasmussen
K.V. Rasmussen
09:15 Jul-13-2005
Re: Ultrasonic angle beam formula for cylindrical forgings.
----------- Start Original Message -----------
: I would like to know the formula for proper angle use to impinge id of thick wallcylindrical forgings/pipe.
------------ End Original Message ------------

Many years ago I found some useful formulas in Materials Evaluation No. 43, November 1985, page 1489.
This was published under the heading 'NDT Solutions' by Jim Johnson.

If you are interested contact me directly via e-mail and I can send you a PDF copy.

N.DHANASEKARAN
R & D
WELDING RESEARCH INSTITUTE, BHEL, India, Joined Jun 2005, 3

N.DHANASEKARAN

R & D
WELDING RESEARCH INSTITUTE, BHEL,
India,
Joined Jun 2005
3
01:42 Jul-15-2005
Re: Ultrasonic angle beam formula for cylindrical forgings.

Hello!
Well known formula to calculate the penetrated depth (d) by angle beam (¥è), knowing outer radius(r) of the pipe is given below:

d = r (1- sin¥è)

By substituting the thickness of the pipe in place of "d", for full penetration by the sound beam, one can calculate the angle required.
For example , 25 mm thick pipe of radius 100 mm will require 45 degree angle probe to detect inner notch in the ref.pipe.
N.DHANASEKARAN

-

---------- Start Original Message -----------
: : I would like to know the formula for proper angle use to impinge id of thick wallcylindrical forgings/pipe.
: Sin¨ª 1 = (ID/OD) * Sin¨ª2
: SIN¨ª2 being the desired angle on the ID surface.
: Sorry misprint.
------------ End Original Message ------------

Product Spotlight

#### FAAST-PA! OEM Patented phased Array for high speed UT inspection

Multiangle, Multifocus, Multifrequency, Multibeam. Instead of stacking UT electronics and having m
...
any PA probes, FAAST-PA is able to transmit all delay laws within ONE single shot in Real time.
>

#### Immersion systems

ScanMaster ultrasonic immersion systems are designed for high throughput, multi shift operation in a
...
n industrial or lab environment. These fully integrated systems provide various scanning configurations and incorporate conventional and phased arrays technologies to support diverse applications, such as inspection of disks, bars, shafts, billets and plates. All of ScanMaster immersion systems are built from high accuracy scanning frames allowing for scanning of complex parts and include a multi-channel ultrasonic instrument with exceptional performance. The systems are approved by all major manufacturers for C-scan inspection of jet engine forged discs. Together with a comprehensive set of software modules these flexible series of systems provide the customer with the best price performance solutions.
>

#### Echomac® PA Tube Weld Tester

The new robotic Echomac® PA TW is an operator friendly test system for detecting weld zone defects
...
in tube and pipe caused by scarfing or welding processes. Electronic scanning ensures that the entire weld zone is covered without the need for mechanical movement or operator adjustment.
>

#### Eddy Current inspection : RotoETscan Rotating Head

The rotating head RotoETscan detects the presence of longitudinal surface or sub surface defects a
...
t high speed. The rotating head is often installed directly on the production line. It is generally dedicated to the control of long products such as tubes, bars and wires, made of ferrous or non-ferrous material. It can also be used for inspection of small parts (billets).
>

Share...
We use technical and analytics cookies to ensure that we will give you the best experience of our website - More Info
Accept
top
this is debug window