where expertise comes together - since 1996 -

# The Largest Open Access Portal of Nondestructive Testing (NDT)

Conference Proceedings, Articles, News, Exhibition, Forum, Network and more

where expertise comes together
- since 1996 -
 747 views
Technical Discussions
Roberto Carlos Giacchetta
Roberto Carlos Giacchetta
05:24 Sep-04-2006
TOFD Hyperbolic cursor

Does anyone know how to draw the Hyperbolic cursor to minimize the error in the depth calculation from a TOFD inspection?

thanks Roberto from IAI CSIC Spain

David Mackintosh
Engineering,
Acuren Group Inc., Canada, Joined Feb 2011, 85

David Mackintosh

Engineering,
Acuren Group Inc.,
Joined Feb 2011
85
00:00 Sep-06-2006
Re: TOFD Hyperbolic cursor

----------- Start Original Message -----------
: Does anyone know how to draw the Hyperbolic cursor to minimize the error in the depth calculation from a TOFD inspection?
: thanks Roberto from IAI CSIC Spain
------------ End Original Message ------------

The equation, if I have uploaded it properly, should appear above. The equation is based on both the defect and the send-receive pair centered on the weld.
C = wave velocity
t = time
y = scan position
d = depth of defect (point-scatterer)
Transmitter position is S

Source: Engineering Applications of Ultrasonic Time-of-Flight Diffraction, 2nd ed., J. P. Charlesworth and J. A. G. Temple, Research Studies Press, 2002, sec 2.3.4 page 39

(Please read the book yourself before you base any important work on this information!)

David Mackintosh
Engineering,
Acuren Group Inc., Canada, Joined Feb 2011, 85

David Mackintosh

Engineering,
Acuren Group Inc.,
Joined Feb 2011
85
02:10 Sep-06-2006
Re: TOFD Hyperbolic cursor
----------- Start Original Message -----------
: Does anyone know how to draw the Hyperbolic cursor to minimize the error in the depth calculation from a TOFD inspection?
: thanks Roberto from IAI CSIC Spain
------------ End Original Message ------------

Back-up to picture in previous response:
Equation is
(C^2 * t^2)/(4 * k^2)  y^2 / k^2 = 1,
where k^2 = S^2 + d^2

Product Spotlight

#### FMC/TFM

Next generation for Phased Array UT is here now with FMC/TFM! Have higher resolution imaging, impr
...
oved signal to noise ratio, characterize, size and analyze defects better with access to several wave mode views and save raw FMC data for higher quality analysis.  Some of the benefits are:
• Beautiful Image! Easier to understand what you're looking at
• Completely focused in entire image or volume
• Much easier to define setups before inspection
• Easier to decipher geometry echoes from real defects
• Oriented defects (e.g. cracks) are imaged better
• See image from different wave modes from one FMC inspection
• FMC data can be reprocessed/analyzed without going back to the field
>

#### FD800 Bench Top Flaw Detectors

The bench-top FD800 flaw detector range combines state-of-the-art flaw detection with advanced mater
...
ial thickness capabilities. Designed for use in the laboratory these gauges are the tool you need for all your flaw detecting needs.
>

#### Eddy Current inspection : RotoETscan Rotating Head

The rotating head RotoETscan detects the presence of longitudinal surface or sub surface defects a
...
t high speed. The rotating head is often installed directly on the production line. It is generally dedicated to the control of long products such as tubes, bars and wires, made of ferrous or non-ferrous material. It can also be used for inspection of small parts (billets).
>

#### Compact NDT inspection-heads for measurements with active thermography

The compact inspection head is suitable for thermographic ndt tasks. The uncooled infrared camera
...
is specially developed for NDI-tasks and offers a thermal sensitivity until now known only from thermal imagers with cooled detector. All required components and functions are integrated into the inspection-head. You will only need an ethernet cable to connect the sensor with the evaluation system.
>

Share...
We use technical and analytics cookies to ensure that we will give you the best experience of our website - More Info
Accept
top
this is debug window