Re: PHASED ARRAY EFFECTIVENESS ----------- Start Original Message -----------
: We paln to buy a PA equipment with 16:64 or 16:128 configurations for using in a production (GAS) platform offshore for corrosion analysis of equipments as well as thickness readings in piping. Can some one advise the PA system's effectiveness or is it worth to buy a PA system for this purpose?wHAT PROBES WE NEED TO DO CORROSION MAPPING ?Do we need TOFD scanner also?
------------ End Original Message ------------
As always in phased arrays, the best probe depends on your specific application. If you are looking for large area coverage in a short time, you could use a 128 element array. The problems with this array are largely from coupling; the array is long, and maintaining coupling may be difficult. You will also need to use a wedge to minimize wear on the probe face, and for a delay line. Normally, we recommend pumping water as the couplant (an aquarium pump normally will do), since it is more difficult to get consistent coupling with smeared-on couplant. However, the ultrasonic coverage is not so good as with smaller arrays with smaller elements.
If you have a particularly rough surface, a smaller array may work better. Also, if you are looking for very small defects, e.g. pinhole corrosion, you may be better off with a smaller array, e.g. 64 elements (also with wedge/delay line) to give you a better pulse density. This assumes that the material is OK for ultrasonics, e.g. carbon steel, not some exotic material.
So, there is no simple answer, but it depends on your surface, the component for access, the speed of scanning required, and the defects. Hope this helps, and keep in touch.
As for a TOFD scanner, maybe and maybe not. It depends on your defects and component. if you want rapid area scanning, TOFD is OK, but you will probably get better and easier analysis using PA. I would try PA first.
NDT Master Lecturer
In the program both university professors and practitioners will give lectures, which guarantees the
oretical depth and practical inside. Academic Director: Prof. Dr. Christian Boller
The following lecturers are not complete: Prof. Tadeusz Stepinski, Prof. Wieslaw Staszewski, Prof. Frank Walther, Prof. Giovanni Bruno, Prof. Gerd Dobmann, Prof. Philippe Guy
TSC Amigo2 - ACFM technology has developed a solid reputation for
accurately detecting and sizing
surface-breaking cracks through
and coatings. As the industry demands increased performance in
signal quality, and portability, it’s time for an evolution. It’s
Mentor UT – Ultrasonic Phased Array Flaw Detector
With Mentor UT, you get an accessible and efficient inspection experience. Create user-defined menus
and workflows (“apps”) with GE’s desktop software Mentor Create to ensure consistency, even for the most complex inspections, every time. Mentor UT is the first UT device to easily allow wireless connectivity and live streaming, so you get second opinions when you need them most—in real-time.