|Stevie Jones |
Face plate thickness
Sorry to come late, could I maybe sneak another
We all know that the best face plate for a transducer
is 1/4 wave thick and its impedance is the mean (either
arithmetic or geometric, depends on the paper) of the
transducer and target impedances.
My question is, what about when either you don't know
what the impedance of the target is, or you cannot
control the impedance of the face that well? Both
apply to soft-faced contact NDT probes. How thick
should the faceplate be then?
|Robert A. Day |
Milky Way Jewels, USA, Joined Nov 1998, 40
Re: Face plate thickness Steve -
The 1/4 wavelength is only for continuous wave sound. Usually the thickness
is a little less for pulsed transducers and in some cases you are better
off with a long delay line. Ideal thickness is at best a compromise. One
way to improve on the compromise is to use two layers one of higher
impedance and one lower in which both are less than 1/4 wavelength. The
actual thickness is usually arrived at empirically in either case.
The case of an unknown impedance is usually handled by just picking
something that works well in general. The types of layers used for contact
transducers for steel usually are poor matches. Use of plastic for example
is popular but clearly doesn't meet the arithmetric or geometric mean
criteria at all. Another popular material is aluminum oxide which matches
the crystal better than either mean. Wear is the criteria being used to
justify aluminum oxide.
Not knowing the impedance is puzzling. Do you mean there is one and you
just don't know it? Or will it vary from part to part or within the part?
Even in these cases you should have some bounds?
In most cases you should be able to do well with any good quality contact
transducer. If much "scrubbing" is involved I would recommend the aluminum
oxide types because they wear much better. The only exception being
plastics where you may want to use a plastic faced probe because wear is
not a problem or the transducer face may wear the part.
: Sorry to come late, could I maybe sneak another
: question in?
: We all know that the best face plate for a transducer
: is 1/4 wave thick and its impedance is the mean (either
: arithmetic or geometric, depends on the paper) of the
: transducer and target impedances.
: My question is, what about when either you don't know
: what the impedance of the target is, or you cannot
: control the impedance of the face that well? Both
: apply to soft-faced contact NDT probes. How thick
: should the faceplate be then?
|Steve Jones |
Re: Face plate thickness Rocky,
: Steve -
: Not knowing the impedance is puzzling. Do you mean there is one and you
: just don't know it? Or will it vary from part to part or within the part?
: Even in these cases you should have some bounds?
I was thinking of general purpose use, where the material could be steel,
or aluminium, or maybe plastic, or ... and you don't want to have hundreds of
pounds tied up in transducers.
: In most cases you should be able to do well with any good quality contact
: transducer. If much "scrubbing" is involved I would recommend the aluminum
OK, you can get by with just about anything. I was wondering if the
academic community had any suggestions for more than
just getting by. It seems not, which is understandable.
: Second Sound
Ultrasonic tomograph for imaging of concrete structures А1040 MIRA
Applicable for concrete inspection allowing imaging of the internal
structure of objects from conc
rete, reinforced concrete, different
stones. The operation applies pulse-echo technique at one-side access
to the object. The instrument is feasible for concrete inspection for
searching conduct ducts, conduits, detection of foreign inclusions,
holes, honeycombing, cracks and other concrete defects.
IRIS 9000Plus - Introducing the next generation of heat exchanger inspection.
Representing the seventh generation of the IRIS system, the IRIS 9000 Plus has nearly 200 years of c
ombined field inspection experience incorporated in its design. This experience combined with a strong commitment to quality and a history of innovation has made Iris Inspection Services® the undisputed leader in IRIS technology.
IntraPhase Athena Phased Array System
The Athena Phased Array system, manufactured by WesDyne NDE Products & Technology, consists of a pha
sed array acquisition system and PC running IntraSpect software. A PC is used to perform acquisition, analysis and storage of the data. System hardware is capable of operating up to four data sets with any combination of phased array or conventional UT probes. NOW AVAILABLE IN 64-64 CONFIGURATION.