I want to know that are there any special codes and standard for Corrosion mapping or we have to just use the Asme Sec V article 4 for Manual UT, Is there any standard calibration block for Corrosion mapping,
How about if we can make one of our own calibration blocks with FBH of 1 mm & 2 mm dia with the different depth simulating the different percentage loss. for example for 10 mm plate 1 mm & 2 mm dia FBH with different depths from 50 % to 90 % of the thickness. Can we make this type of calibration block and prove the sensitivity.
Mark Nel Sales, Application support Technology Design Ltd., United Kingdom, Joined Jul 2005, 58
Mark Nel Sales, Application support Technology Design Ltd., United Kingdom, Joined Jul 2005 58
01:57 May-28-2009 Re: Corrosion Mapping Calibration Block & StandardsIn Reply to Abbas Bombaywala at 14:55 May-25-2009 (Opening).
Abbas, As corrosion mapping is an in service inspection and is usually requested as part of the wider corrosion control effort, it is to my knowledge rarely approached from a strict regime controlled by a specification but rather a procedure or technique is established based on the corrosion mapping equipment manufacturers particular method of applying this procedure in his hardware/software and configuring these to conform with the corrosiom department/engineers requirements. This department or individual quite often even looks to the service provider to supply the procedure for the task. There may even be several alternatives that you could choose from to best complete the task. Although, I guess there's nothing wrong with making a block with artificial defects to qualify the procedure, remember that all you really are doing is measuring thickness and translating these wall thickness readings into a colour-graphic image that provides an excellent means of identifying trends. The actual data may also be interrogated in minute detail using the available software tools (cursors etc.) to provide specific depth readings of each recorded area (pixel). I suggest that you contact the equipment vendor, as suggested by a previous respondent, to get a step-by-step guide on how to configure system for corrosion mapping.
SIUI’s newly launched Smartor is a combination of ultrasonic testing
and ultrasonic thickness me
●Compact size: 198 (W)* 128 (H) *520 (D) mm
●0.9kg only with battery
●5.7" LCD with high resolution 640×480 pixels
●Multiple conventional UT functions
●Smart Test Wizard
The all-digital Novascope 6000 is a portable, ultra-high precision
thickness gauge for high-speed
Novascope 6000 has unmatched capabilities and unique features
•Superior Resolution with high contrast, high-speed color RF
•High pulser voltage
•Real-time video output
•Increased internal/external data storage
•Programmable SetUp features
•Battery & AC Powered