best method for ndt cast iron thickness?
What is the best method for NDT of cast iron? Application is paper machine dryer cylinders. We have historically just used standard pulse echo probes, then went to an automated time of flight (TOF, also known as hartford wedge I believe?) method due to inaccuracies found with . We have found the TOF to be very difficult to schedule as it scans the length of the roll by running on a rail with position feedback etc. Processing is done through special software by contractors.
We would like to go back to a manual measurement but want the accuracy of the TOF.
Are there any standard instruments out there that will work with a TOF probe.
Is there a smart way a standard instrument can be used, i.e. just collect both signals ( surface and thickness) and use the surface actual/measured factor to adjust the thickness measurement?
Are there other NDT tecnologies out there for measureing materials such as cast iron.
Its been a while since I have worked with this gear so go easy on me jargon wise. I have just been asked to reseach possible alternatives.
Any help much appreciated.
Norske Skog Tasman,
NDT Inspector, API-510, 570, 653
MISTRAS, USA, Joined Jan 2009, 148
Re: best method for ndt cast iron thickness? In Reply to Mike Plant at 02:35 Jul-22-2009 (Opening).
We've had good-to-excellent results with conventional UT-straightbeam. Our setup consists of a 1.0 MHz 1/2" to 1" dia. 'ducer on a top-quality flawmeter. Panametrics Epoch 3 or 4, Krautkramer 50 or 60 series. Digital thickness-only meters don't work, dual transducers don't work, medium frequency 'ducers work poorly.
Set the meter to measuring peak-to-peak, NOT 'main bang' to peak. Calibrate on a sample of the material, single point cal. works well [assuming the cal block and the dryer are at the same temperature]. This is because you are measuring peak-to-peak on a linearly accurate meter, so the only calibration item is the actual velocity of your material.
What you are measuring on Peak-to-Peak is the time of flight of the UT pulse between the ID & OD of the drum. The 'quality' of the signal is easily verified, as a good signal has 4 to 7 reflection peaks on the meter screen, in a steadily decreasing series. If you cannot see 3 peaks, the data is probably not valid. Might be a 'tracking' error of the crawler, or not enough couplant.
One B-scan of one cylinder should only take about 1-1/2 to 2 minutes, just long enough for the linear encoder on the crawler 'buggy' to drive from one end to the other. The advantage of the B-scan over several spot measurements is that you now have a profile drawing of the interior of your drum, and can exaggerate the height measurements to easily show banding or dishing of the drum ID.
HD-CR 35 NDT Computed Radiography System
Portable high-resolution CR scanner for all radiography applications - weld testing, profile images
and aerospace. No matter what type of radiographic testing you are performing, the unique TreFoc Technology of the HD-CR 35 NDT imaging plate scanner always guarantees the highest image quality.
NOVO Armor 15 & NOVO Armor 22
The Armor Kit Contains the NOVO Armor, which provides additional mechanical protection to the NOVO 1
5WN & NOVO 22WN Detectors, the Armor Stand and a traveling soft cover.
- Newest shock absorbent technology case
- Water resistant design
- Supports wired & wireless communication
- Multiple positioning options
- Tripod connection using the Built-in 1/4” threads
- Simple Detector battery replacement
35 years of VOGT Ultrasonics!
This year we are celebrating a company anniversary! 35 years have passed since
we first took our t
est device and swung the probe on behalf of safety. But our
ultrasonic know-how goes back many years..
High Resolution Industrial CT System for Small/Medium Size Parts Inspection
The YXLON FF35 CT computed tomography system is designed to achieve extremely precise inspection res
ults for a wide range of applications. Available in a single or dual tube configuration, it is perfect for very small to medium size parts inspection in the automotive, electronics, aviation, and material science industries.