Material Attenution while using DGS
I have been a member to this forum for several months now and this is my first post. I enjoy reading and absorbing the intelligence that is out there.
I have been using a USN58L for about 3 years and have the DGS option. I have used the DGS option for several techniques and would like to know how to properly use the Material Attenuation function since I work with many exotic grades, primarily consisiting of Monel, Inconel, Copper, and Duplex, but not limited to common grades like 4340, 8630, all forms of stainless steel. I want to make sure that this function is set up properly to get a consistent test and proper test.
On another note. This option has several transducers programmed into with specific curves for these transducers. When not using one of these transducers what is the proper way to set-up for the transducer being used,...ie GE RHP 1" 2.25mhz standard contact transducer commonly used.
Thanks for all your help,
NDT Inspector, API-510, 570, 653
MISTRAS, USA, Joined Jan 2009, 148
Re: Material Attenution while using DGS In Reply to DAC at 18:40 Dec-29-2009 (Opening).
DGS is basically "DAC Curves from the Factory". It is a pre-calibrated attenuation [& beam spread] curve for that exact model instrument, using that exact transducer, on one specific calibration material. DGS eliminates the need to have calibration blocks, and allows you to use a calibration performed by an experienced factory tech, under ideal conditions.
To use DGS on an exotic material, the factory has to have programed in a curve for that material. To use DGS on any material, not only does a curve have to have been pre-programed into your instrument, but you have to use the exact transducer that the programming tells you to use. And the 'ducer has to be in 'as-new' condition. "DAC Curves for Dummies".
DGS seems to be common in Europe. Typically in the USA, a set of calibration blocks are used to create a DAC curve for the material to be inspected, using your existing transducer, in whatever condition that 'ducer is in. Thus wear or roughness of the 'ducer face and/or the delay tip are part of the calibration you will use to inspect your parts. Personally, I won't certify my DGS results unless the 'ducer is in pristine condition, and the material listed in the factory calibration is an acoustic match to the material being tested. Yes, it is possible to get fairly close results with a worn DGS 'ducer, but how accurate is 'fairly close'?
Magnetic X-Ray Pipeline Crawler
Zhong Yi brand pipeline crawler is magnetic/micro-video controlled
crawler with DC X ray unit inst
alled for checking welding of pipeline.
Move steadily inside the pipeline 6''-60''diameter with speed of up to
18m/min, Max. moving diatance 5 kilometers and provide the efficient
inspection of the pipeline.
MUSE Mobile Ultrasonic Equipment
The MUSE, a portable ultrasonic imaging system, was developed for in-field inspections of light-weig
ht structures. The MUSE consists of a motor-driven manipulator, a water circulation system for the acoustic coupling and a portable ultrasonic flaw detector (USPC 3010). The MUSE provides images of internal defects (A-, B-,C- and D-scan).
IRIS 9000Plus - Introducing the next generation of heat exchanger inspection.
Representing the seventh generation of the IRIS system, the IRIS 9000 Plus has nearly 200 years of c
ombined field inspection experience incorporated in its design. This experience combined with a strong commitment to quality and a history of innovation has made Iris Inspection Services® the undisputed leader in IRIS technology.
Research and Applications Development For NDT
The Research and Applications Development (RAD) group is a newly formed team
within Acuren dedicat
ed to tackling challenging inspection problems. Our focus
is the development of novel, field deployable, advanced inspection techniques
for use in cases where standard NDT methods are ineffective. We don't wait for
new innovations, we engineer them. From concept to commissioning.