1108 views  Technical Discussions   Tony Martland , Joined , ^{}   03:09 Sep242002 Calculation of particle velocity....please help !!!! Hello Can anybody help me ? I am trying to find out how to calculate the particle velocity of a longitudinal wave in a solid. I am applying the wave by means of a transducer, into mild steel, and I know the wavespeed in the mild steel (5900 m/s) but how do I calculate the actual particle velocity. ANY help would be much appreciated. Thanks Tony
 
   Paul A. Meyer R & D, GE Inspection Technologies, USA, Joined Nov 1998, ^{47}   06:15 Sep242002 Re: Calculation of particle velocity....please help !!!! Hi Tony, The particle velocity is proportional to the ratio of the sound pressure to the acoustic impedance. You know the acoustic impedance but the sound pressure is a function of the transducer design, the driving pulse, and the coupling conditions, making it very difficult to determine. If you have access to a KLM transducer model, you can perhaps estimate the sound pressure in the steel. Paul : Hello . : Can anybody help me ? I am trying to find out how to calculate the particle velocity of a longitudinal wave in a solid. I am applying the wave by means of a transducer, into mild steel, and I know the wavespeed in the mild steel (5900 m/s) but how do I calculate the actual particle velocity. . : ANY help would be much appreciated. . : Thanks . : Tony .
 
   luis marques refinery inspector petrogal, Portugal, Joined Jul 2002, ^{5}   06:50 Sep242002 Re: Calculation of particle velocity....please help !!!! It depends on what kind of apparatus you are using. Simple digital ultrasonic equipment such as a Krautkramer DM4 will do what you want. With this equipment if you do not know the sound velocity in the medium you are scanning, you just need a calibration block of a known thickness and same material, you want to scan. In these calibration conditions the equipment will display automatically the sound velocity.Regards Luis
 
   Joe Buckley, Sonatest Plc Consultant, ASNT LIII, Honorary Secretary of BINDT Level X NDT, BINDT, United Kingdom, Joined Oct 1999, ^{519}   07:21 Sep242002 Re: Calculation of particle velocity....please help !!!! This is one that could 'get extremely silly' remember that:1. The net value of the particle velocity is zero  They dont move, just oscillate about a fixed point (thats why its solid) 2. The bulk properties of the material (Youngs modulus etc) are not necessarily appropriate things to use when you get to the atomic level. 3. You cant necessarily assume a sine wave or anything like that, so building up a theoretical model will be quite complex  May be worth asking exactly what is it you 'really' want to know? Joe : Hi Tony, : The particle velocity is proportional to the ratio of the sound pressure to the acoustic impedance. You know the acoustic impedance but the sound pressure is a function of the transducer design, the driving pulse, and the coupling conditions, making it very difficult to determine. If you have access to a KLM transducer model, you can perhaps estimate the sound pressure in the steel. : Paul : : Hello : . : : Can anybody help me ? I am trying to find out how to calculate the particle velocity of a longitudinal wave in a solid. I am applying the wave by means of a transducer, into mild steel, and I know the wavespeed in the mild steel (5900 m/s) but how do I calculate the actual particle velocity. : . : : ANY help would be much appreciated. : . : : Thanks : . : : Tony : . .
 
   Pierre Marty R & D Saab Aeronautic, Sweden, Joined Jul 2002, ^{5}   04:02 Sep262002 Re: Calculation of particle velocity....please help !!!! HI TonyA possible way is to use a dynamic Finite Element Model. It is not so difficult to estimate the profile of the sound pressure at the interface betrween the transducer and the sample, so that you can model the system with a good accuracy. If you have any mean (laser probe ?9 to measure the displacement at the outer surface of the transducer and if you assume a good coupling between the transducer and the sample than the FE method will give you a fair estimation of the particle displacement in the bulk of the steel sample. The limits of such aproach are those of the FE method and of your own assumptions in term of displacement profile and magnitude at the surface of the transducer. I beleive commercial FE programs such as ANSYS, PZFLEX can do that. IF you need contact me for more info on that subject. I did some similar models myself with another FE code. Pierre : Hello . : Can anybody help me ? I am trying to find out how to calculate the particle velocity of a longitudinal wave in a solid. I am applying the wave by means of a transducer, into mild steel, and I know the wavespeed in the mild steel (5900 m/s) but how do I calculate the actual particle velocity. . : ANY help would be much appreciated. . : Thanks . : Tony .
 
  ISAFE3 Intrinsically Safe Sensor System ISAFE3 intrinsically safe sensor system of Vallen Systeme is especially targeted at the petrochemica ... l  as well as oil and gas transportation industry. The sensor system is designed for permanent monitoring or periodic inspection tasks. Sensors are available for different AEfrequency ranges optimized for corrosion and fatigue crack detection and other applications.
The ISAFE 3 sensor system consists of an AEsensor (model ISAS3) certified according to ATEX/IEC for installation in zone 0, gas group IIC, IP68, 20 to +60 °C, and a signal isolator (model SISO3) certified for installation in zone 2. An ISAS3 sensor can be mounted in atmosphere or submerged, e.g. in water or crude oil. It is supported by mounting tools for temporary (magnets) or permanent (welded) installation.
ISAFE3 supports automatic sensor coupling test and can be used with any AE signal processor supporting 28V supply at 90 mA peak, e.g. Vallen Systeme ASIP2/A.
> CIVA 2017 UT ModuleCIVA NDE Simulation Software is the world leader of NDT
Simulation.
The UT simulation Module incl ... udes:
 "Beam computation": Beam propagation simulation
 "Inspection Simulation": Beam interaction with flaws or
specimens
The user can simulate a whole inspection process (pulse echo,
tandem or TOFD) with a wide range of probes (conventional, Phased
arrays or EMAT), components, and flaws.
> 

We use technical and analytics cookies to ensure that we will give you the best experience of our website  More Info
