where expertise comes together - since 1996 -

# The Largest Open Access Portal of Nondestructive Testing (NDT)

Conference Proceedings, Articles, News, Exhibition, Forum, Network and more

where expertise comes together
- since 1996 -

 302 views
Technical Discussions
Terry Oldberg
Engineering, Mechanical Electrical Nuclear Software
Consultant, USA, Joined Oct 1999, 42

Terry Oldberg

Engineering, Mechanical Electrical Nuclear Software
Consultant,
USA,
Joined Oct 1999
42
08:15 Oct-26-2002
Protocol for Measurement of POD?

A member of the nde.net forum wrote to me by email:

Interesting to read your comments on POD over the years on the net. Do you have such a thing as an IDIOTS guide (PROTOCOL), on how you feel is the best way to conduct a POD exercise.

I responsed:

Thanks for the interesting question.

There is a situation in which a protocol used frequently in science is available. The situation is that a partition of the material that is inspected by a test, "Partition A," is defined and is fixed as a property of this test. The purpose of the test is to measure the value of one or more properties of each element of this partition. A test is referenced that establishes the true value of each measured property, "Ground Truth".

Partition A forms a legitimate, statistical population for purposes of measuring the reliability of the test. Now, define a partition of the values of one of the measured properties into two parts, "partition B". Define one of these parts as "positive" and the other as "negative". All of the permutations of measured and true values can be now represented by events conventionally termed "true negative," "false negative," "true positive," "false positive," Estimate the false positive and false negative error probabilities by post inspection random sampling in the population, followed by establishment of the ground truth in the sample.

This protocol is inconsistent with your desire to estimate the POD. What you might be able to do is adopt this protocol and use it to measure a property that contains nearly the same information as the POD.

For example, if it is possible to place the boundaries between the elements of Partition A in places where it is extremely unlikely for a flaw to cross a boundary, a probability that captures similar information can be estimated for a test that measures the size of the largest flaw in each element of Partition A; sizes above "z" reside in one of the two elements of Partition B and the remaining sizes reside in the other element. With "positive" defined as values at or above z and z set at a "relatively low value," the probability of a true positive contains information that would be almost identical to the information in the POD, for many purposes; by "relatively low value," I mean a value that is a bit below the value at which the largest flaw becomes capable of propagating to failure.

Product Spotlight

#### Scatter Correction for Industrial Cone-Beam Computed Tomography (CBCT)

Download our white paper on Scatter Correction for Industrial Cone-Beam Computed Tomography (CBCT)
...
Using 3D VSHARP, a fast GPU-Based Linear Boltzmann Transport Equation Solver here: http://bit.ly/2Y0t78P
>

#### Ultrasonic Probe Recharacterization Service

NDT Systems offers a comprehensive Aftercare and Recharacterization Service for all our ultrasonic
...
probes. The Recharacterization Service is fully compliant with International ASTM E1065 Standard Guide (and other applicable standards) and offers complete documentation, traceable to the ASTM E1065 Standard. For more details or to schedule Recharacterization Services contact ndtsales@ndtsystems.com
>

#### TraiNDE UT

TraiNDE UT is a virtual tool associated with a signal database which simulates real inspection con
...
ditions for numerous applications (Type A/V1 block, DAC block, welds and plates).
>

#### OPBOX with standard software is able to do all types of inspections and measurements: flaw detection in welds and materials, scanning of objects, testing composite materials, forged and moulded pieces, many UT inspections, measurements of properties of ma

Typical applications: UT measurements with pulse technique, Measurement of thicknesses also at hig
...
h temperatures, Measurements of properties of materials, including fluids and gases We are delivering a standard version of the software (for any Microsoft Windows up to 10 x64 with Microsoft Hardware certification report Approved) and for special needs: SDK with ready to use examples for LabView, MATLAB x64, C++ wrapper for dll, Python and Linux., and also low-level description of how to control our devices directly from any USB tools
>

Share...
We use technical and analytics cookies to ensure that we will give you the best experience of our website - More Info
Accept
top
this is debug window