where expertise comes together - since 1996 -

# The Largest Open Access Portal of Nondestructive Testing (NDT)

Conference Proceedings, Articles, News, Exhibition, Forum, Network and more

where expertise comes together
- since 1996 -

 881 views
Technical Discussions
zaki
, Joined ,

zaki

,
Joined

09:19 Nov-27-2002
IR Thermography

Just for curiosity..

In Thermography method, measurement appear in Temperature (oC), but theoritically the measurement is performed in term od total receiving radiation power (W). Is there any relationship between temperature vs. radiation..

Alberto Monici
Engineering, Leak Detection Service - Thermography Service -
ETS Sistemi Industriali, Italy, Joined Aug 2002, 17

Alberto Monici

Engineering, Leak Detection Service - Thermography Service -
ETS Sistemi Industriali,
Italy,
Joined Aug 2002
17
09:55 Nov-27-2002
Re: IR Thermography
Dear Mr.Zaki
For a black-body the relationship is that gived by Plank's law, in wich:
W= (2*pi*h*c^2)/((lambda^5)(e^(hc/lambdakT)-1))
in which W is the spectral emissivity [Watt/m^2*meter] and T is the black-body temperature in Kelvin.
The intergal of the Plack formula in all the lambda possible give the Total Energy Emission (Stefan-Boltzmann law):
Wt=k*T^4
where k is the Stefan-Boltzmann constant.
For a grey-Body (like that normally measured in nature), the emissivity of the body, its spectral transmittance and the spectral absorbing coefficient give a total emittance below then a black-body. So the formula is: Wt=E*k*T^4, where E is the ratio between emittance of grey and black bodies.
I Hope to be exhaustive for your couriosity. Don't exitate to contatct me for any other question.

:
: Just for curiosity..
.
: In Thermography method, measurement appear in Temperature (oC), but theoritically the measurement is performed in term od total receiving radiation power (W). Is there any relationship between temperature vs. radiation..
.

Robert Paynter
R & D,
U of Oxford, United Kingdom, Joined Jun 2001, 14

Robert Paynter

R & D,
U of Oxford,
United Kingdom,
Joined Jun 2001
14
01:33 Dec-02-2002
Re: IR Thermography
Alberto Monici wrote:
: Dear Mr.Zaki
: For a black-body the relationship is that gived by Plank's law ...

:: Just for curiosity..
:
: In Thermography method, measurement appear in Temperature (oC), but theoritically the measurement is performed in term od total receiving radiation power (W). Is there any relationship between temperature vs. radiation..
: .
.
Alberto Monici gave the total, but most sensors have a limited band of detection and you need to allow for that. CMT (Cadmium/Mercury/Telluride) is the favourite at typical ambient temperatures because it's response ends up being almost linear in this range. InSb (Indium Antimonide) can do these low temperatures but is much better (more linear) at higher (600K +).
They are well behaved responses so will be calibrate within the camera.

As a user you need to take care that the emission of your observed surfaces is taken into account - some are unexpectedly good with IR eg. white snow.

Robert

Product Spotlight

#### High-performance Linear Phased Array Probes

Available to order from stock in a range of 5MHz – 7.5MHz and from 16 to 64 elements. Designed w
...
ith piezo-composite elements, Phoenix phased array probes provide high-resolution imaging to maximise sensitivity; accurate ultrasonic detection and sizing of defects in welds; and effective corrosion mapping. Housed in a rugged stainless steel case for on-site industrial NDT applications.
>

#### Sci Aps Z-Series Portable Handheld Analysers

The world’s only handheld analyzer that measures carbon content in stainless (yes even L-grades),s
...
teels, and cast irons. Also accepted for low Si analysis for sulfidic corrosion analysis, and is widely used in the power industry for Cr analysis, for flow accelerated corrosion applications.
>

#### CIVA 2020 UT Module

CIVA NDE Simulation Software is the world leader of NDT Simulation. The UT simulation Module incl
...
udes: - "Beam computation": Beam propagation simulation - "Inspection Simulation": Beam interaction with flaws or specimens The user can simulate a whole inspection process (pulse echo, tandem or TOFD) with a wide range of probes (conventional, Phased- arrays or EMAT), components, and flaws.
>

#### Extended Range Variable Wall® Piping Calibration Blocks for Longseam Inspection

This set of two blocks has been specially designed for situations where ultrasonic examination tak
...
es place circumferentially or “around the curve” for applications such as inspection of longseam welds in piping and vessels. Jointly designed by PH Tool and Holloway NDT, the blocks feature side-drilled hole reflectors positioned within novel compound-curves, each with multiple thickness steps. The unique design results in a light and compact form factor, with the largest block weighing just over 15 lbs and spanning just 16 ¾”. Coverage is provided for nominal pipe sizes from 6” to 10” in the small block and 12” to a whopping 56” in the large block. The compound OD curves and multiple wall thicknesses actually exceed ASME requirements, providing exceptional signal response and measurement accuracy. Properly managing beam divergence off the ID and stabilizing beam skew due to probe rocking is essential to maintaining a consistent calibration when inspecting lengthwise on curved surfaces.
>

Share...
We use technical and analytics cookies to ensure that we will give you the best experience of our website - More Info
Accept
top
this is debug window