where expertise comes together - since 1996 -

# The Largest Open Access Portal of Nondestructive Testing (NDT)

Conference Proceedings, Articles, News, Exhibition, Forum, Network and more

where expertise comes together
- since 1996 -
 565 views
Technical Discussions
Terry White
Terry White
04:29 Dec-17-2002
U/T rep rate

We test nickel alloy billet material using basic pulse echo techniques. our method spec calls for a minimum pulse repetition rate of 500PPS. We often experience "wraparound" at this rate and have to lower the rep rate to eliminate the unwanted signals. My question is, why is it so important to maintain that rep rate? What harm could dropping it to 450 - 400 PPS do? Max surface scanning speed is 10 lin. inches per/sec. Transducer is 10 or 5MHz. Beamwidth as measured at .150" metal travel on a #1 FBH cal block is @ .040".

Udo Schlengermann
Consultant, -
Standards Consulting, Germany, Joined Nov 1998, 182

Udo Schlengermann

Consultant, -
Standards Consulting,
Germany,
Joined Nov 1998
182
06:46 Dec-17-2002
Re: U/T rep rate
Hello Terry,

If your scanning speed v is 10"/s and your beam width b is only 0.040", i.e. per shot you cover only 0,040", you need a pulse repetition rate for complete coverage of
frep = v/b.
For your example that means frep = (10"/0.04"s)=250Hz at least.
If you want a factor n for redundancy, i.e. n pulses per beam diameter b, the equation goes to
frep = n v/b.
In your case n=2 means frep = 2 10"/.040"s) = 500 Hz.

You are correct, that this repetition frequency may be too high for low attenuating materials as nickel based alloys are.
The only solution then is to look for the maximum repetition rate fmax you can work with (static), and for scanning to lower the scanning speed until you get full coverage including a safety factor n:
vmax = (b fmax/n)
If you get fmax = 50 Hz, with n = 2 you will get the maximum allowable scanning speed vmax = (0.04" 50/2s)= = 1"/s only.
May be, for lower repetition rates you need a digital ultrasonic instrument, because they have separate screen refreshing rates independent from pulse repetion frequency, whereas with analog instruments the screen refreshing rate is the same as the pulse repetition frequency and for low rates you will not see a constant screen display.

Kind regards
Udo Schlengermann
Agfa NDT GmbH, Huerth, Germany
+++

: We test nickel alloy billet material using basic pulse echo techniques. our method spec calls for a minimum pulse repetition rate of 500PPS. We often experience "wraparound" at this rate and have to lower the rep rate to eliminate the unwanted signals. My question is, why is it so important to maintain that rep rate? What harm could dropping it to 450 - 400 PPS do? Max surface scanning speed is 10 lin. inches per/sec. Transducer is 10 or 5MHz. Beamwidth as measured at .150" metal travel on a #1 FBH cal block is @ .040".
.

Product Spotlight

#### On-Demand Webinar: Advanced Ultrasonic Techniques for HTHA Detection & Assessment

In the Oil and Gas industry, High Temperature Hydrogen Attack (HTHA) can occur in steels operating a
...
t high temperature, usually above 205°C (400°F). Early stages of HTHA are very difficult to detect. Join us for a webinar presented by Guy Maes, Zetec Sales Engineer Director. The webinar will demonstrate how advanced ultrasonic techniques are deployed for efficient and reliable HTHA inspections.
>

#### FD800 Bench Top Flaw Detectors

The bench-top FD800 flaw detector range combines state-of-the-art flaw detection with advanced mater
...
ial thickness capabilities. Designed for use in the laboratory these gauges are the tool you need for all your flaw detecting needs.
>

#### AIS229 - Multipurpose Real Time System

Latest standard & automatic real time system developed by Balteau. The AIS229 has been designed to
...
do series inspection in a wide variety of industry. Composed of a shielded cabinet, 5 axis manipulator, x-ray generator and tubehead from 160kV to 225kV, a fl at panel & much more, the AIS229 is most certainly one of the most multipurpose RTR system available on the market.
>

#### NDT.net launches mobile-friendly design

NDT.net has revamped its website providing a mobile-friendly design.The front page received a comp
...
letely new design and all other sections are now reacting responsively on mobile devices. This has been a major step to make our website more user- friendly.
>

Share...
We use technical and analytics cookies to ensure that we will give you the best experience of our website - More Info
Accept
top
this is debug window