|K Fosha |
Taking thickness readings on FRP material.
I would like to find out if there is such a thing as a, UT transducer, that is able to take thickness readings of FRP pipe material. (* FRP = Fiber Reinforced Plastic *)
I don't think that a transducer would be able to use sound to obtain a thickness reading due to the amout of glass fiber in the material. But then again, I could be completely wrong and need to be shown the light.
|Neil Burleigh |
Krautkramer Australia Pty Ltd, Australia, Joined Dec 2002, 152
Re: Taking thickness readings on FRP material. : I would like to find out if there is such a thing as a, UT transducer, that is able to take thickness readings of FRP pipe material. (* FRP = Fiber Reinforced Plastic *)
: I don't think that a transducer would be able to use sound to obtain a thickness reading due to the amout of glass fiber in the material. But then again, I could be completely wrong and need to be shown the light.
: K Fosha
Over the years I have had a number of experiences with FRP. Krautkramer once had a thickness gauge the CL202F which uses the ZIP zero interface probe, it is now no longer available. This probe works well as it is a 0.5MHz probe with a special delay block to reduce the interface echo, ( as the name implies). Details can be found on the KK website about ZIP probes.
Recently we have been using the piezo-composite probes for testing FRP pipe and rubber conveyor belt with great success. Try the K1SC, MSEK2 and MK1S
|Taha R Sheikh|
Re: Taking thickness readings on FRP material. In Reply to Neil Burleigh at 06:28 Dec-18-2002 .
i just read ur reply below.
unfortuatly i cant get any of the below mentioned products in KK website.
Please send me the link or advice me accordingly for thickness montioring of FRP ducts!
|Sudheer Jai Krishnan|
Re: Taking thickness readings on FRP material. In Reply to Taha R Sheikh at 18:01 Feb-28-2013 .
Please let us know your min and maximum thickness. Have you tried a 45MG or a 38DL Plus for the thickness measurements? If not please see the link below and We have given solutions for similar kind of application to many of our customers. The highlighting feature is that ,the 38DLP has got both precision thickness measurements and corrosion measurements such that you can have one equipment for your different requirements
Kindly let me know if i can be of any assistance to you.
Sudheer Jai Krishnan
Royal Canadian Air Force, Canada, Joined Oct 2012, 273
Re: Taking thickness readings on FRP material. In Reply to Sudheer Jai Krishnan at 21:01 Feb-28-2013 .
Another option is the new Zip line of probes available through General Electric... Check online to find a representative near you...
retired, USA, Joined Nov 1998, 390
Re: Taking thickness readings on FRP material. In Reply to K Fosha at 02:14 Dec-17-2002 (Opening).
Yes, there are transducers suitable for fiberglass testing. For more than 25 years Panametrics and now Olympus NDT has offered a specialized 500 KHz delay line transducer, the M2008, that is optimized for thickness measurement and detection of laminar flaws in fiberglass up to 100 mm or thicker. It uses an impedance matched delay line for good coupling and minimal interface echo interference. It can be used with high penetration thickness gages or with a flaw detector. The application note in the link below references fiberglass boat hulls, but the principle is applicable to fiberglass pipes as well.
We (and other major manufacturers as well) also offer a number of other low frequency transducers that can be effectively used on fiberglass in various thickness ranges. If your pipe radius is less than approximately 150 mm we would first try a smaller diameter transducer as an alternative, such as a 1 MHz or 2.25 MHz, 12.5 mm diameter broadband contact transducer.
35 years of VOGT Ultrasonics!
This year we are celebrating a company anniversary! 35 years have passed since
we first took our t
est device and swung the probe on behalf of safety. But our
ultrasonic know-how goes back many years..
Mentor UT – Ultrasonic Phased Array Flaw Detector
With Mentor UT, you get an accessible and efficient inspection experience. Create user-defined menus
and workflows (“apps”) with GE’s desktop software Mentor Create to ensure consistency, even for the most complex inspections, every time. Mentor UT is the first UT device to easily allow wireless connectivity and live streaming, so you get second opinions when you need them most—in real-time.
Next generation for Phased Array UT is here now with FMC/TFM! Have higher
resolution imaging, impr
oved signal to noise ratio, characterize, size and analyze
defects better with access to several wave mode views and save raw FMC data for
higher quality analysis. Some of the benefits are:
- Beautiful Image! Easier to understand what you're looking at
- Completely focused in entire image or volume
- Much easier to define setups before inspection
- Easier to decipher geometry echoes from real defects
- Oriented defects (e.g. cracks) are imaged better
- See image from different wave modes from one FMC inspection
- FMC data can be reprocessed/analyzed without going back to the field
NDT.net launches mobile-friendly design
NDT.net has revamped its website providing a mobile-friendly
design.The front page received a comp
letely new design and all
other sections are now reacting responsively on mobile devices.
This has been a major step to make our website more user-