Could anyone suggest a way to solve the Lamb wave dispersion equations? I want to plot the dispersion curves for a number of symmetric and antisymmetric modes, the equations, however, need a numerical evaluation.

Michael Conry Student University College Dublin, Ireland, Joined May 2002, ^{3}

Michael Conry Student University College Dublin, Ireland, Joined May 2002 ^{3}

07:43 Dec-02-2003 Re: Lamb wave dispersion equations ----------- Start Original Message ----------- : Hello, : Could anyone suggest a way to solve the Lamb wave dispersion equations? I : want to plot the dispersion curves for a number of symmetric and : antisymmetric modes, the equations, however, need a numerical evaluation. : Thank you in advance, : Vasilis Protopappas ------------ End Original Message ------------

Plotting the curves can be tricky. The easiest way is as follows:

Divide the frequency/phase-velocity (f/c) space up into small increments.

At each point within this space calculate the value of the characteristic equation F(f,c)

Then do a contour plot of the resulting numbers, including just a single contour (at F(f,c)=0)

This is easy to do with Matlab, and I have also done it with Gnuplot and Python.

This is not an efficient way to plot the curves (it requires a lot of calculations) but even for a multilayered plate I found it took a pretty short length of time to prepare a nice dispersion plot.

Doing the job in a more rigorous way requires either that you use curve following algorithms to plot the curves (e.g. find the solutions for a particular value of phase velocity, then incrementally reduce the phase velocity and find solutions at each subsequent point). A difficulty in this case can be correct handling of the crossovers of the curves.

00:29 Nov-25-2006 Re: Lamb wave dispersion equations Hello I would really be appreciate if somebody ask me why in the lamb wave dispersion curves, just A0 mode close to zero as frequency be zero? best regards khash khash1979@yahoo.com

Teletest Focus+ electronics have superior capabilities than rivals on
the market. Beyond the usual

...

test features, Focus+ has 24 transmit
channels and 24 receive channels, with an additional on-board
switching capabilities. The instrument's frequency range is 10–300
kHz.

For highly attenuating materials, the performance of the system is critical. The ultrasonic sensors,

...

the scanning area and the system settings should be flexibly adapted to the test task and the material.
These high expectations are met with the new and modular testing system SONOAIR. With the world’s first air-coupled phased-array UT inspection system SONOAIR we developed a technology that works with up to 4 transmitter and receiver channels with freely configurable square wave burst transmitters as wells as low noise receiving amplifiers.