where expertise comes together - since 1996 -

# The Largest Open Access Portal of Nondestructive Testing (NDT)

Conference Proceedings, Articles, News, Exhibition, Forum, Network and more

where expertise comes together
- since 1996 -

 1740 views
Technical Discussions
Don
Don
06:58 Mar-04-2004
Calculation of velocity of surface waves

Does anybody know the equation for calculating the surface (Rayleigh) wave velocity, given the material properties, modulus of elasticity, poisson's ratio density etc?

I have seen rough calculation as 0.92*ShearVelocity, and a more accurate result using a calculator at www.eng.vt.edu/fluids/msc/jscalc/wscac02.htm but do not know the equations hidden inside it. Any ideas, Don

Paul A. Meyer
R & D,
GE Inspection Technologies, USA, Joined Nov 1998, 47

Paul A. Meyer

R & D,
GE Inspection Technologies,
USA,
Joined Nov 1998
47
02:55 Mar-04-2004
Re: Calculation of velocity of surface waves
In ULTRASONIC WAVES IN SOLID MEDIA, Joseph Rose references an approximate solution by Viktorov. The ratio of the surface wave speed to the shear wave speed (Cr/Ct)= (0.87 + 1.12*nu)/(1+nu) where nu is Poisson's ratio. For nu = 0.29 the ratio is 0.92 as you stated.
Paul

----------- Start Original Message -----------
: Does anybody know the equation for calculating the surface (Rayleigh) wave velocity, given the material properties, modulus of elasticity, poisson's ratio density etc?
: I have seen rough calculation as 0.92*ShearVelocity, and a more accurate result using a calculator at www.eng.vt.edu/fluids/msc/jscalc/wscac02.htm but do not know the equations hidden inside it. Any ideas, Don
:
------------ End Original Message ------------

Neil Burleigh
Sales
Krautkramer Australia Pty Ltd, Australia, Joined Dec 2002, 159

Neil Burleigh

Sales
Krautkramer Australia Pty Ltd,
Australia,
Joined Dec 2002
159
00:51 Mar-05-2004
Re: Calculation of velocity of surface waves
----------- Start Original Message -----------
: Does anybody know the equation for calculating the surface (Rayleigh) wave velocity, given the material properties, modulus of elasticity, poisson's ratio density etc?
: I have seen rough calculation as 0.92*ShearVelocity, and a more accurate result using a calculator at www.eng.vt.edu/fluids/msc/jscalc/wscac02.htm but do not know the equations hidden inside it. Any ideas, Don
:
------------ End Original Message ------------

Don,
There is a formula for the calculation of velocity of surface waves given on pg 22 of "The Krautkramer Booklet" (The blue book)It looks like it is the same formula as the one that Paul Meyer offered but it has not presented as a ratio. If you don't have a "blue" book contact your local Krautkramer Rep.

Regards
Neil Burleigh
P.S. The blue book now has a white cover.

Product Spotlight

#### Conformable wedge transducer

The conformability is obtained with a flexible membrane filled with water between the transducer and
...
the inspected component. The coupling between the membrane and the component requires a small quantity of water or couplant. The conformable wedge combines the acoustic performance of immersion technique with good coupling and low attenuation.
>

#### Research Center IDEKO develops an ultrasonic train wheel inspection system for CAF

The Basque Research centre IK4-IDEKO has developed a state-of-the-art ultrasonic inspection system f
...
or the train wheel. This system secures a sound condition of train wheels and is thus a significant contribution to rail transport safety. The device was delivered to CAF recently and its use allows the manufacturer to become an approved supplier of rolling stock in Italy, as it meets the demanding homologation standards of this country. \\\\r\\\\n
>

#### NDT.net launches mobile-friendly design

NDT.net has revamped its website providing a mobile-friendly design.The front page received a comp
...
letely new design and all other sections are now reacting responsively on mobile devices. This has been a major step to make our website more user- friendly.
>

#### HD-CR 35 NDT Computed Radiography System

Portable high-resolution CR scanner for all radiography applications - weld testing, profile images
...
and aerospace. No matter what type of radiographic testing you are performing, the unique TreFoc Technology of the HD-CR 35 NDT imaging plate scanner always guarantees the highest image quality.
>

Share...
We use technical and analytics cookies to ensure that we will give you the best experience of our website - More Info
Accept
top
this is debug window