|Wilby Daniels |
Bonded ultrasonic transducers and methods for making
I have heard that it is the type backing material that is critical in the making of a low noise ultrasonic transducer. I have used some different types of epoxies from hysol but I have not had good results. Can anyone assist me with techniques or ideas for other materials to try? I understand too that matching impedance at or near crystal is important.
|Carlos Valdecantos |
MTORRES, Ing., Spain, Joined May 2001, 9
Re: Bonded ultrasonic transducers and methods for making Backing must match the crystal impedance (usually high) but, at the same time, must strongly atenuate any pulse going backwards. Unfortunately no single material shows these two features, oppossing each other in some way, at the required level. A classical solution makes use of epoxi resin (low viscosity, not too fast)+metallic powder+vermiculite (globular mica). Tungsten is ideal because its very high density and its powder is commercially available. Cork particles are also useful instead of vermiculite. Appliying such a mixture to the crystal and leaving it to settle leads to a composite backing with high impedance (tungsten rich) close to crystal and high attenuation (vermiculite rich) at the back surface. Centrifugation is sometimes used to get higher impedance at the interface.
Good luck. And happy New Year.
|Paul A. Meyer |
R & D,
GE Inspection Technologies, USA, Joined Nov 1998, 47
Re: Bonded ultrasonic transducers and methods for making Hello Wilby,
When you excite a piezoceramic, you store energy in it in the form of resonant vibrations. The energy is dissipated as heat, or as it is coupled into adjacent materials. To remove the energy quickly, the adjacent materials should be close to the acoustic impedance of the piezoceramic (ceramics are ~ 20 to 35 Mrayles). Epoxies have impedances about 3 to 5 Mrayles and, therefore, do not damp the ceramic very well. Adding dense particles to the epoxy can increase the impedance, tungsten particles work well. Additionally, the particles scatter the energy in the mixture reducing reflections back into the ceramic. The books "Ultrasonic Testing of Materials" by Krautkramer, and "Ultrasonic Transducers for Nondestructive Testing" by Silk cover these topics.
I hope this helps.
----------- Start Original Message -----------
: I have heard that it is the type backing material that is critical in the making of a low noise ultrasonic transducer. I have used some different types of epoxies from hysol but I have not had good results. Can anyone assist me with techniques or ideas for other materials to try? I understand too that matching impedance at or near crystal is important.
------------ End Original Message ------------
AIS229 - Multipurpose Real Time System
Latest standard & automatic real time
system developed by Balteau. The
AIS229 has been designed to
inspection in a wide variety of industry.
Composed of a shielded cabinet, 5
axis manipulator, x-ray generator and
tubehead from 160kV to 225kV, a fl at
panel & much more, the AIS229 is most
certainly one of the most multipurpose
RTR system available on the market.
Lyft™: Pulsed Eddy Current Reinvented
PEC Reinvented—CUI Programs Redefined
Corrosion under insulation (CUI) is possibly the greatest u
asset integrity problem in the industry. Current methods for
measuring wall thickness with liftoff, without removing insulation,
all have severe limitations. Eddyfi introduces Lyft — a reinvented,
high-performance pulsed eddy current (PEC) solution. The patent-
pending system features a state-of-the-art portable instrument, real-
time C-scan imaging, fast data acquisition with grid-mapping and
dynamic scanning modes, and flexibility with long cables. It can also
scan through thick metal and insulation, as well as aluminum,
stainless steel, and galvanized steel weather jackets. Who else but
Eddyfi to reinvent an eddy current technique and redefine CUI
programs. Got Lyft?
Teletest Focus+ electronics have superior capabilities than rivals on
the market. Beyond the usual
test features, Focus+ has 24 transmit
channels and 24 receive channels, with an additional on-board
switching capabilities. The instrument's frequency range is 10–300
EKOSCAN Phased Array
In order to always fit your needs, EKOSCAN can manufacture any type of UT transducer, either convent
ional or Phased Array.
As an ISO 9001: 2015 certified company, EKOSCAN is extremely careful as far a material selection and manufacturing processes are concerned. Our probes guarantee our customers the benefits of latest innovations regarding piezo-composite, backing, impedance adaptation layer, etc.
Specific probes for hostile environment: high temperature, high pressure, corrosive environment,etc.
Specific probes designed to fit your specific application: optimization of every parameter to guarantee you the best detection.